How to solve the problem that there is no result, the percentage of Sulfur, after analyzing by 5E-S3200?
1. Check there is flow showed in flowmeter.
2. If there is flow, then check the gas leak in the system. From the electrolytic cell to the end of pump.
For the cell, need to check the O-ring is installed properly,
For the drying tube, check the O-ring at the top when you changing the silica gel.
And need to make sure all the pipe connection is ok.